Fluke 435-II Power Quality and Energy Analyzer Repair
Will not power on
Only beeps when power button is pressed
Battery won't charge
Won't read voltage or amperage
Inaccurate frequency, power factor, or harmonics measurements
Input waveform signal won't trigger or hold steady
Keypad buttons not working or responding to presses
Broken input BNC or banana terminal jacks
Broken SD memory card socket
"No SD Card" message when card is installed
USB port broken or not working
Typical Repair Cost Range: $795 - $2,150*
* Based on previous repairs; actual cost will be quoted after unit evaluation and may be more or less
Tips & Pointers
If the analyzer will not turn on when a known good battery is installed and/or the unit is plugged into the charger adapter, it is possible the keypad contact foil could be defective. Replace the foil and try again. If the problem persists, further diagnosis and repair is necessary.
If the analyzer only beeps when the power button is pressed, there is an internal problem with the digital processor or related circuitry and the unit must be serviced.
Erratic or unstable waveforms or inaccurate or no channel readings may indicate one or more of the input terminal jacks (bananas or BNCs) are cracked or broken. The input terminals are part of a larger assembly and the unit must be professionally serviced to correct the issue.
Our repair service is fast, easy, and professional. For over a decade, ScopemeterRepair.com has repaired hundreds of Fluke units for customers all around the world.
How To Request Repair
Tell us what is wrong with the instrument and who to contact by doing one of the following:
Ship the unit to us using your choice of carrier. Our shipping address depends on what model you are sending and will be shown on the confirmation e-mail.
We will diagnose the problem and send you a formal quote for repair.
Repair Request Form
* If your instrument's model is not listed, please note it above and CONTACT US for support options.
Apt, suite, etc.
After submitting the form above, you will receive an e-mail confirmation with shipping instructions. Ship your unit to our receiving address:
ScopemeterRepair.com, LLC 5641 Broyles Rd Hartville, MO 65667 United States
No payment is due until all services are completed and the unit is ready to ship. We accept all major credit cards on our website and over the phone, PayPal, and checks or money orders.
Turnaround time is typically 5-10 business days but is workload-dependent and may vary. If you require immediate service, CONTACT US to let us know your requirements. We also offer 24-, 48-, and 72-hour turnaround times for most models for an additional fee.