Fluke 435 Three-Phase Power Quality Analyzer Repair
Will not power on
Only beeps when power button is pressed
Battery won't charge
Display has bands or vertical lines
Won't read voltage or amperage
Inaccurate frequency, power factor, or harmonics measurements
Input waveform signal won't trigger or hold steady
Keypad buttons not working or responding to presses
Broken input BNC or banana terminal jacks
Typical Repair Cost Range: $445 - $975*
* Based on previous repairs; actual cost will be quoted after unit evaluation and may be more or less
Tips & Pointers
If the analyzer will not turn on when a known good battery is installed and/or the unit is plugged into the charger adapter, it is possible the keypad contact foil could be defective. Replace the foil and try again. If the problem persists, further diagnosis and repair is necessary.
If the analyzer turns on briefly then immediately shuts off, the battery may be bad and must be replaced. Disconnect the battery and try powering the analyzer from the charger alone. If it turns on, then the battery is likely at fault. If it still shuts off, further diagnosis and repair is necessary.
Erratic or unstable waveforms or inaccurate or no channel readings may indicate one or more of the input terminal jacks (bananas or BNCs) are cracked or broken. The input terminals are part of a larger assembly and the unit must be professionally serviced to correct the issue.
Our repair service is fast, easy, and professional. For over a decade, ScopemeterRepair.com has repaired hundreds of Fluke units for customers all around the world.
How To Request Repair
Tell us what is wrong with the instrument and who to contact by doing one of the following:
Ship the unit to us using your choice of carrier. Our shipping address depends on what model you are sending and will be shown on the confirmation e-mail.
We will diagnose the problem and send you a formal quote for repair.
Repair Request Form
* If your instrument's model is not listed, please note it above and CONTACT US for support options.
Apt, suite, etc.
After submitting the form above, you will receive an e-mail confirmation with shipping instructions. Ship your unit to the address shown in the e-mail. Multimeters will ship to our Eastern lab while Scopemeters, calibrators, analyzers, and process meters will ship to our Midwest lab.
No payment is due until all services are completed and the unit is ready to ship. We accept all major credit cards on our website and over the phone, PayPal, and checks or money orders.
Turnaround time is typically 5-10 business days but is workload-dependent and may vary. If you require immediate service, CONTACT US to let us know your requirements. We also offer 24-, 48-, and 72-hour turnaround times for most models for an additional fee.